In this episode Shahriar repairs an Agilent PSA Series Spectrum Analyzer. The instrument generates many errors during self-alignment and produces no measurements below 3.2GHz. The block diagram of the unit is thoroughly presented and various possible failure points are considered. Based on the observation of the noise floor, the most likely cause is the second LO module. The measurement of the LO power indicates that the second LO power is fall below nominal.
The second LO signal (at 3.6GHz) is generated by a DRO PLL module which is locked to 600MHz. The PLL is functional, however the output power is below -20dBm. Teardown of the module reveals a simple design with a doubler and filter. The filter (which is discolored) shows a huge loss likely caused by RF losses due to trace degradation. Thinning the traces provides some improvement and the remaining loss is compensated with a two stage RF amplifier. With this modifications, the instrument’s alignment errors are partially resolved.
The next problem is with the input attenuator at the 20dB range. This problem is simply resolved by providing lubrication on the attenuator solenoids after disassembly. The instrument is then used for various measurements to verify its correct operation.
The Signal Path