In this episode Shahriar takes a detailed look at an Agilent (Keysight) E4407B ESA-E Spectrum Analyzer. The instruments reports an “Unlock LO” error message which prevents internal alignments from being performed. Before the repair process can begin, the firmware of the instrument must be updated. The firmware upgrade requires the installation of additional flash memory ICs which is presented in the video. The main processor board with its various components are also described in the video.
The complete block diagram of the spectrum analyzer is presented with a detailed look at the principle of operation of the instrument. All the main functional blocks such as the YIG Oscillator, Phase Detector, LO Path, LO Amplifier, Charge Pump, Pre-tune segments, digital circuits, IF and LO paths are analyzed both in schematic and on the PCB. Several potential faults are investigated and measured.
The Signal Path